All product names are trademarks of their respective companies. See the ACUSON Cypress System Operator’s Manual for a list of WARNINGS. Acuson Cypress system manual. Somkid Sridaromont; 10 videos How to use the Acuson Cypress Ultrasound System. by EchoVTorials. Learn how to use the Cypress ultrasound system, learn the Cypress in about an hour with.
|Published (Last):||14 August 2007|
|PDF File Size:||9.41 Mb|
|ePub File Size:||15.89 Mb|
|Price:||Free* [*Free Regsitration Required]|
Siemens ACUSON Cypress Operator’s Manual
Advertise Increase Your Brand Awareness. External Devices And Connections Caching Of Modality Worklist Results M-mode Soft Windows Adjusts the sweep speed of the spectral display.
Colorize Click [ ] to start the time counter. Acuson Cypress Service Manual January 29, Turn the knob until the desired label is displayed.
Make sure Cypress system is Connect the keyboard cable to the external keyboard connection on the Cypress cypres see Figure Mnaual have an account? On the right side of the screen click the Enabled box. Dicom Storage Commitment Explanation Of Patient Menu Buttons It includes information about: Siemens makes no claims regarding the safety or efficacy of ultrasound contrast agents. Page This chapter includes information about: Modality Worklist Details screen.
Stops the timer clock. Accept When the box is checked, the Cypress system will query Storage server to verify that all images in the study DICOM Commitment have been properly received and stored.
SIEMENS ACUSON CYPRESS OPERATOR’S MANUAL Pdf Download.
Type the desired estimate into the edit box. Turn the Soft Window knob until the best image is playing. See the sections in this chapter for specific information about each scanning mode. Page Turn the knob to select a number.
The timer can also be started manually by clicking ] and selecting Start. Page – Making Corrections to the Trace and Area Storing A User-defined Preset Page In Freeze Mode: For example, Sweep 1 is a 1-second sweep; Sweep 2 is a 2-second sweep; etc.
Page Patient ID elements,andDICOM respectively, will still contain the valid patient identifying information for the image. Connecting An Amnual Printer W ycpress the transverse axis. Vascular Calculation References All Modalities to search for patients scheduled on any imaging modality. Moving the Cypress System within a Building Make sure that the transducer is securely placed in the transducer holder.
Overview Of Cardiac Measurements See Appendix B for information about ink-jet printers.
Siemens Acuson Cypress Plus ACUSONSystem Operator’s Manual-English | PDF | MedWrench
Connecting An External Monitor Peripheral Device Connections Acuson Cypress Service Manual September 17, This will provide lower default values and maunal access to indices for neonatal cephalic studies. When the screen is recorded on videotape and played back, portions of the screen—at the bottom and right side—are not displayed.
Always check the condition of the cleaning brush by opening the shutter of the manuak cleaner. Page After cleaning, be sure that fluff from the swab is not left on the sur- face of the ball-cage or optical lens.